Find the most up-to-date version of DIN at Engineering DIN Field procedure for precision testing of surveying instruments; general information. DIN Field procedure for precision testing of surveying instruments; gyroscopes. standard by Deutsches Institut Fur Normung E.V.
|Published (Last):||20 January 2007|
|PDF File Size:||13.34 Mb|
|ePub File Size:||18.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
August 3, 4: Repeat the above 4 times on different days.
That’s the article I would have pointed you at, and realistically all you need to know, unless you’re doing some sort of higher order stuff. Wow this is awesome thanks! It is ddin test done with equipment that negates the personal pointing error of the instrument man. Target 1 will always average out to zero.
Add value or move along. Error is always present in all measurements. Or in German for that matter at a reasonable price? Recheck the instrument level If the instrument has moved, check the angle just measured. Yes, the most fundamental and easy way to upgrade angular accuracy is just to use well-designed targets that are up to the task and use methods xin actually center them over the ground marks they are nominally occupying. My wording was confusing.
Di the standard error or the standard deviation for session 1. What that meant was that you could measure the same direction times and each direction as read by the instrument would have the identical error generated by the circle reading system. Please note, that a DIN is specific to one instrument. We think you have liked this presentation.
ADJUSTMENT COMPUTATIONS – ppt video online download
Applied to methods and instruments used to attain a high order of accuracy. The first one is for the “testing theodolites” and the second is for levels. Statistical Treatment of Data Significant 81723 And by probably, I mean definitely.
What indicates to me that it’s for all instruments is that the title says “theodolites”. If you look at the precision specs of a Trimble instrument, the angular precisions are usually given using DIN standards.
Square the Sum of the difference between the measurement and mean Divided by the number of repetitions minus one. Please Login or Register. Taking readings with the above steps Compute the average direction of each set for each target. Published by Kiara Dinn Modified over 3 years ago.
Too many surveyors no longer do angle sets.
If a company makes 1, 2, 3, 5, and 7″ guns, it’s not because they manufactured them differently, it’s because the 7″ gun didn’t meet the 1″ spec. Work of the hardest kind. In other words, “your mileage may vary”. Error tends to be random Random error- If the bubble of a theodolite were to remain off center by the same amount during the entire angle-measuring process at a station, the resulting error would be systematic.
Log in or sign up in seconds. Share buttons are a little bit lower. Accuracy relates to the quality of the result obtained when compared to the standard. Auth with social network: Now, if a surveyor is using poorly defined targets in poor seeing condititions, sure, all bets are off. Each source produce a small amount of random error.
Land Surveying Community Forums
See below for screen shot of my data sheet of my instrument. Did you by chance cover the steps for zenith angle, levels, or EDM in your lecture? Submit a new text post. Amount and quality of data acquired GPS signal strength and continuity Ionosphere and troposphere conditions 118723 site stability, obstructions, and multipath. DIN is a procedure for determining the accuracy, or standard error, of the mean pair of Direct and Reverse directions to a single target.
August 3, 5: Before hitting submit, check that your content doesn’t violate the rules below: